Bad header dates in firmware 1.2.2 #
|Open Acoustics Devices
The date found in the header will be incorrect because the UTC to local time adjustment is miscalculated.
See this commit for details.
Fixed in the next firmware 1.3.0.
The problem existed between
Some times in the header of the files are basically inaccurate (± 3600s) but only IF a fractional timezone is set and only for fimware version 1.2.2. The filename datestamp should be unaffected.
There is no way to detect the difference in the AudioMoth comment format between versions 1.2.2 and 1.3.0.
Effects of the problem on common tools #
Some packages will report incorrect times.
EMU’s metadata will automatically adjust dates when it finds discrepencies and emit a warning.